The case works with the majority of Insulin Pumps.
Important Note regarding Insulin Pumps and Aquapac's case
Aquapac have commissioned research by Imperial College London into the use of the 158 (Insulin Pump case) and 558 (Connected Electronics case) after receiving a large number of customer enquiries regarding their suitablity for use with insulin pumps. Results show that, in some circumstances, insulin flow rate can be reduced by the case. In particular, low flow rates are impeded, and as such Aquapac can make no claim regarding the suitablity of these cases with insulin pumps. Use of these cases with insulin pumps is entirely at the discretion of the user.
Aquapac has been advised to recommend that users of Aquapac cases with insulin pumps regularly check their blood sugar levels.
The tests were conducted by Imperial College of Science, Technology and Medicine using a Roche Accu-Check Spirit insulin pump at a temperature of 20C at flow rates of 2U/hr and 10U/hr. The amounts of insulin delivered over 60 minutes observed were as follows:
2U/hr not in a case - 0.0131g (flow rate of 100%)
2U/hr in Aquapac 150 - 0.0106g (flow rate of 81%)
2U/hr in Aquapac 150 - 0.0056g (2nd test model)(flow rate of 42%)
2U/hr in Aquapac 550 - 0.0121g (flow rate 92%)
10U/hr not in case - 0.0864g (flow rate of 100%)
10U/hr in Aquapac 150 - 0.0878g (flow rate of 100%)
10U/hr in Aquapac 550 - 0.0903g (flow rate of 100%)
21st August 2006
|Weight||Case: 1.7oz / 47g
Belt: 2.1oz / 59gm
|Add to Compare|
Review by Lynn, Berkshire, UK, added 6/26/12: "Great"My daughter went on an activity holiday (first time since she was diagnosed 18 months ago). My daughter had a brilliant time, thanks to your waterproof case she was able to swim plus loads of muddy activities wearing her pump. There are hundreds of patients children and adults going on holidays and they have to keep their pump dry. I hope you have plenty in stock!!